Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||27 September 2010|
|PDF File Size:||17.94 Mb|
|ePub File Size:||18.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: Supply circuits and output circuits are designed to meet the standards of overvoltage classes III and II, respectively.
KROHNE IFC D Calibration – Micro Precision
The operator is, howeverobligated to take note of these sections in the. Signal cable to be permanently laid. Tell kgohne what’s missing. Page 10 3 Start-up 3. Power the unit, and the flowmeter will immediately start process flow measurement.
KROHNE IFC 010 Converter Short EN User Manual
Page 15 Function 3. Don’t see a manual you are looking for? Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter igc Cable entries 2 Electrical connection of outputs 2. Switches, counters, displa ys and recorders.
Lev el measuring instruments. Tell us about it. Page 14 Function 1. All operating data have been factory set in accordance with your specifications. Page 2 Contents 1 1. Your manual failed to upload Passive mode External power source required for operation of the outputs.
V ariable area flowmeters.
KROHNE Aquaflux K/D/6 with Signalkonverter IFC D Unused | eBay
Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2. Connect shields via stranded drain wires.
These concise instructions do not include the following: P Pulse output overranged.
What’s missing? Tell us about it.
If installed and operated in accordance with these operating instructions, it will rarely present any problems. Page 3 1 Electrical connection: The flowmeter housings protecting the electronic equipment from dust and moisture kohne always be kept closed.
Vo r tex flowmeters. Or point us to the URL where the manual is located.
Do not cross or loop cables inside the terminal compartment. Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal krohe 6 Cable entries 2 Electrical connection of outputs 2.
Underwater and underground installation possible. Page 11 Standard factory settings Function 1.