Find great deals for Agilent / HP A GPIB DC Power Supply VDC a W. Shop with confidence on eBay!. HP / Agilent A is Watt, single output, series pass regulated programmable power suppliy optimized for systems applications. HP A Price. There is a new tool in the lab: An HP A power supply. It might be a a bit dated, but it has some pretty good and useful specs ( V, 5 A).
|Published (Last):||15 May 2017|
|PDF File Size:||19.90 Mb|
|ePub File Size:||3.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
HP Models AA have a built-in selftest capability.
Highly recommended for a heavy usage with different settings. Given my previous work on fan speed control, this is 6362a challenge too good to pass on. This means the item has been inspected, np, and repaired to full working order and is in excellent condition.
I use it to inject specific voltages in various electronic circuits, mostly RF type, to trouble shoot issues in ham radios. The ground seems to be connected to mains earth by about ohms. It provides high performance output with a full set of programmable features. Save on Adjustable Power Supplies Trending price is based on prices over last 90 days. These are late 90s models from HP, capable of V and A.
The negative current setting tracks the 6632 positive current CC setting. Skip to main content.
People who bought this also bought. See details for full description. HP Ph transformer, oh this is what makes the thing so heavy…. If you don’t mind the weight, size, and noise this a great value and its always better to buy used: Offer AccuSource Electronics is always happy to consider reasonable yp on our test instruments.
Subscribe to RSS feed. Please let us know which of these you would be willing to consider for the price discount. The original Papst fan still sounds like a jet engine though, it is probably worth looking into a replacement.
Agilent / HP 6632A
6362a More information to come about this project in the future. Additional Info Please provide any additional information which could help us determine whether we can meet your requirements for this offer.
Message me if you are interested in the project and might want to buy a PCB! Leave a Reply Click here to cancel reply. Great and reliable power supply Great power supply. I have two on my work bench with front terminals option installed.
This site uses Akismet to reduce spam. Sorry, your blog cannot share posts by email.
HP A 20 Volt 5 Amp DC Power Supply | eBay
Unable to Process Request Your offer was received but we are unable to process it for the following reason: We hope to earn your business.
HP A output shorted 6623a a 10ohm resistor, voltage set to 5V, current set to mA, fast mode. Thank you for allowing us the opportunity to assist with your test equipment needs.
I even tried the overcurrent protection feature but it is equally slow; it just disables the output when the CC-mode kicks in. 66322a item doesn’t belong on this page.
Hmmm… Worked a treat — almost better than the original! This item may or may not be in original packaging.
HP Agilent Keysight, A
I ended up taking it apart so I could mount binding posts on the front. Your email address will not be published. It 6632a quieter, but does not provide as much airflow. Notify me of new posts by email. This feature will compensate for load lead IR drops of up to 2 volts per lead. Which means you can use one as a programmable load. I used the kind found in mains power centrals in residential housing, the copper is so thick it actually takes some effort to bend it.
Call us if you do not find what 662a are looking for Unable to Process Offer Your offer was received but we are unable to process it for the following reason: Both the voltage and current output can be programmed for either CV or CC operation.
HP A sink as well as source current. They also have a really loud fan gotta do something about that… and push around W of heat out when idle.
TICS International Ltd., UK
HP A output shorted with a 10ohm resistor, voltage set to 5V, current set to mA, normal mode. However, setting the output to 15 V, the supply shows Turning the supply on it worked a treat, but it hl LOUD!
It might be within spec, but there does seem to be a ca mV negative offset in the output. The Overvoltage protection includes an SCR crowbar. HP A power supplies. Make sure to 6362a proper, heavy duty wires between the PCB and the jacks, in theory 5A at 20V could be flowing through there.