Gonotec – Osmomat The OSMOMAT is the standard system for the determination of the total osmolality in aqueous solutions. This system is commonly. Gonotec Gesellschaft für Meß- und Regeltechnik mbH does not assert any claim on trademarks or trade names other than its own. ® OSMOMAT model. Find pricing on new and used Gonotec Osmomat ‘s and other lab products. Connect with a variety of sellers on LabX and buy today.
|Published (Last):||24 August 2015|
|PDF File Size:||10.77 Mb|
|ePub File Size:||19.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
It is simple osmomt operate, very reliable and provides extremely rapid sample handling. Bend it into place with your hand if necessary. If the built-in printer is switched on two values are printed out for a standard calibration solution.
The picture here FUSE small arrow underneath the power cord connector.
If Gonotec is notified within the warranty period of a defect in a hardware product and if the Gonotec warranty applies to the defect, Gonotec shall repair or replace the defective product at its discretion. Gonotec Osmomat Cryoscopic Osometer Price: Osmomag auto Freezing point osmometer.
If you no longer have the original packaging, you odmomat purchase a replacement from Gonotec. REF Article number or order number The following pages provide you with a step-by-step introduction to using, maintaining and servicing the measurement equipment. In these countries, the warranty is only valid in the country in which the product was purchased.
Special version for 3-point calibration Appendix To ensure optimal reproducibility, the thermistor probe must be exactly in the center of the measuring vessel during measurement.
The results can be seen on the digital display after each measurement. A repeat determination using just one sample is not bonotec, as the sample separates during measurement.
Calibration at three points. Replacing an initiation needle. Gonotec Osmomat Cryoscopic Osmometer. Antimicrobials — 20th Annual Scientific Meeting.
Gonotec Osmometer & Chloridmeter
Insert a new roll of paper. The Osmomat auto osmmomat determine the total osmolality of aqueous solutions when small sample volumes are available or a short measuring time is required. Osmolality is an important gnotec of concentration for diluted solutions, such as all bodily fluids.
Description Features Photos 1. The OSMOMAT obtains a reading for the total osmolality by calculating the temperature difference between the freezing point of the water and the reversal point detected while measuring the sample solution.
If the sample consists of a solution, the pure water in the solution crystallizes spontaneously and the substances move to the rest of the solution. Always use a new measuring vessel with fresh distilled water for each new measurement. Correct feed direction for paper roll.
Gonotec Osmomat | Forums | Questions | Discussions | Help | LabWrench
Sections 2 Special Regulations for Active Medical Devices and 3 Medical Devices with Measurement Functions regulate the safety and measurement checks to be performed on medical devices. Pressing this key again toggles through the following values: Please pay special attention to chapter 7. Replacing an initiation needle Step 2 a. RS interface with pin assignment 8.
Gonotec Osmomat 030 Cryoscopic Osmometer
Distributors Find distributors Become distributor. Thermistor with mounting plate Next, insert the plug of the connector cable into the connector and guide the thermistor cable into the groove along the side.
In the idle position of the motor swipe, the end of the needle must not extend below the lower edge of the cooling pin. Mismeasurements that do not produce a result gohotec the digital display and during which the LED for spontaneous crystallization or no crystallization illuminate are not counted.
These checks are mandatory for devices listed in Annexes 1 and 2 of the Medical Devices Operator Ordinance.