The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||3 February 2013|
|PDF File Size:||10.87 Mb|
|ePub File Size:||12.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
The EC restart automatically. Right click and select Unistall for both the COM1 ports. A chassis terminal; a connection to the instrument’s chassis, which includes all exposed metal structure. This document contains proprietary information that is protected by copyright.
The boot option is displayed. Do not replace components with the power cable connected. Remove the two stickers item 1.
Atec Agilent-E5071C User Manual
Factory recovery Returns e0571c contents of the C drive to the factory state. Figure 3 Item 3: The following screen appears when the user backup image was created.
If anything is missing, contact the nearest Agilent Technologies office. Operation of any electrical instrument in such an environment clearly constitutes a safety hazard.
In addition to the Windows operating system, the following settings of the EC are returned to the factory state. NOTE With no operation on the touch screen calibration screen for a preset time, it automatically closes and the previous measurement screen reappears.
A message dialog box appears after the completion of the Recovery Image Setup procedure. Select the D drive to restore, then click Next button.
Inset the key in the lock and turn crockwise. The manual part number changes when extensive technical changes are incorporated. Click Port x Short to start the calibration measurement x denotes the test port to which the standard is connected.
If you press F11 twice or more and a screen which is not mentioned in this manual appears, reboot the instrument and restart the process from the first. The mark x appears also on the upper right, lower left, and lower right. Click Maunal x-y Thru to start the calibration measurement x and y denote the test ports between which the THRU connection is being made.
Select the C drive to restore, then click Next button. In the next dialog box, input agena in the Name box.
Reconfiguring Touch Panel Settings Step 1. Click Port x Load to start the calibration measurement x denotes the test port to which the standard is connected. The printing date changes when a new edition is printed. Figure Recover Factory Backup ecse Step 8.
Agilent Technologies E5071C Manual
EC firmware is executed, then the measurement display appears. If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered. To use this function, you must prepare for recovery in advance. Select the saved backup image file, then click Next button. The System Properties dialog box appears. RestoreSysCorFile dialog box appears Figure In the procedure below, the selected test ports are denoted as x and y. Execute the calibration of the touch screen.
If you use a calibration kit other than a predefined one, you need to define it. The recovery takes a few minutes e5071 on the amount of data.
Atec Agilent-EC User Manual | 60 pages
For more information, see Modifying Calibration Kit Definition. Turn the key to the left until it is locked. It is recommended to save the file to a directory which has enough space like the E drive.
CN or below Step 1.